Our stock products are being manufactured in our plants continuously in order to provide you a shortest lead time as possible.
In the following table you may find typical lead times. They may vary depending on quantity and raw material availability. Some products are having an interchangeable parts.
You will receive an accurate delvery date on your order confirmation after you ordered.
|Optomechanics||Average 2 - 5 weeks*|
|Optics||Average 2 - 5 weeks*|
|Motorized Stages||Average 3 - 6 weeks*|
|System||Average 3 – 6 weeks*|
*This is an average typical lead time. For more details, please ask to: firstname.lastname@example.org
The stock availability for this product is limited at the moment. Please contact the sales team for more information at email@example.com.
Estimated average shipping time is 2 - 10 working days. For urgent shipping please contact the sales team at firstname.lastname@example.org. Extra charges may applies.
At this time pre-manufactured parts are already available for a quick assembly.
The typical lead times is estimated at 1 - 3 weeks after your order, depending on quantity.
You will receive an accurate delivery date on your order confirmation after you ordered.
For more details, please ask to : email@example.com
A beam sampler behaves like a plate beamsplitter, it has the ability to reflect approximately 5.2% of the total beam.
◦Uncoated surface of optical parallels or wedged substrates are used as reflection surfaces. The rear surfaces are coated with multi-layer anti-reflection.
◦These products have transmitted beam deviation and ghosting of the rear surface reflections due to the characteristics of plate beamsplitters.
◦Wedged beam samplers with AR coating on the rear surface should be selected to prevent ghosting.
▶Please contact our Sales Division for customized products. (Customized on size, wavelength or R:T, etc.)
▶For a guarantee in reflected wavefront error or transmitted wavefront error, please contact our Sales Division with your requests.
▶Wedged types are marked with an arrow on the side of the substrate indicating the thickest point of the wedge.
▶The Beam Sampler has a reflectance of 5.2% when the material is BK7 and the input beam is unpolarized or circularly polarized.
▶The transmitted beam deviation of a wedged beamsplitter is larger then with a beamsplitter made of an optical parallel.
▶The amount of beam deviation of a beamsplitter depends on thickness of the substrate and the wavelength of the incident angle of the input beam.
▶Be sure to wear laser safety goggles when checking optical path and adjusting optical axis.
|Coating||Front surface ; Uncoated |
Rear surfaces ; Visble multilayer antireflection coating
|Divergence ratio |
(reflectance : transmittance)
|5 : 95 |
(The average value of the P-Polarization and the S-Polarization)
|Laser Damage Threshold||4J/cm2 (Laser pulse width ; 4ns , Repetition frequency ; 20Hz)|
|Surface Quality |
|Clear aperture||90% of the diameter|