Our stock products are being manufactured in our plants continuously in order to provide you a shortest lead time as possible.
In the following table you may find typical lead times. They may vary depending on quantity and raw material availability. Some products are having an interchangeable parts.
You will receive an accurate delvery date on your order confirmation after you ordered.
|Optomechanics||Average 2 - 5 weeks*|
|Optics||Average 2 - 5 weeks*|
|Motorized Stages||Average 3 - 6 weeks*|
|System||Average 3 – 6 weeks*|
*This is an average typical lead time. For more details, please ask to: firstname.lastname@example.org
The stock availability for this product is limited at the moment. Please contact the sales team for more information at email@example.com.
Estimated average shipping time is 2 - 10 working days. For urgent shipping please contact the sales team at firstname.lastname@example.org. Extra charges may applies.
At this time pre-manufactured parts are already available for a quick assembly.
The typical lead times is estimated at 1 - 3 weeks after your order, depending on quantity.
You will receive an accurate delivery date on your order confirmation after you ordered.
For more details, please ask to : email@example.com
This objective lens can be used for laser machining using pulsed laser of SHG (532nm) YAG laser and FHG (266nm) YAG. Chromatic aberration is suppressed in both the visible and UV laser wavelength, achieving a high transmittance.
◦With its long working distance and field curvature corrected, its natural observation image is obtained to the periphery of the visual field.
◦It is the long working infinity correction function that is used to introduce a laser system and coaxial observation.
◦It is also used for the observation of ultra-violet light.
◦Laser Damage Threshold (Typical) 0.09 J/cm2 (266nm), 0.2J/cm2 (532nm) (Laser pulse width 10ns, repetition frequency 20Hz)
▶Available fixed objective lens holder (LHO-26).
▶When the objective lens is fixed to a 2 axis holder, please consult our Sales Division.
▶▶For laser processing, we offer a dichroic block (DIMC W2041) and for laser unit with coaxial illumination and observation (OUCI-2).
▶Please refer to here for Wavelength Characteristic Data (Reference Data, EXCEL file format).
▶When an objective lens is used in laser processing, use the diameter of the incident beam to extend to a size of half the pupil diameter (1/e2). A small light spot cannot be achieved when the incident beam is too narrow. Please note if there is a laser energy density increase, there will be a high possibility of damage to the objective lens.
▶The surface of an objective lens can be contaminated by debris during processing. To avoid this, please have sufficient working distance (WD) and insert a thin protective glass on the objective.
▶Magnification is the value when using the imaging lens f=200mm. When used in a microscope lens barrel from other manufacturers there may be different magnifications. The actual magnification should be calculated from the ratio of the focal length of the objective lens and the focal length of the imaging lens to verify the focal length of the imaging lens barrel to be used.