Our stock products are being manufactured in our plants continuously in order to provide you a shortest lead time as possible.
In the following table you may find typical lead times. They may vary depending on quantity and raw material availability. Some products are having an interchangeable parts.
You will receive an accurate delvery date on your order confirmation after you ordered.
|Optomechanics||Average 2 - 5 weeks*|
|Optics||Average 2 - 5 weeks*|
|Motorized Stages||Average 3 - 6 weeks*|
|System||Average 3 – 6 weeks*|
*This is an average typical lead time. For more details, please ask to: firstname.lastname@example.org
The stock availability for this product is limited at the moment. Please contact the sales team for more information at email@example.com.
Estimated average shipping time is 2 - 10 working days. For urgent shipping please contact the sales team at firstname.lastname@example.org. Extra charges may applies.
At this time pre-manufactured parts are already available for a quick assembly.
The typical lead times is estimated at 1 - 3 weeks after your order, depending on quantity.
You will receive an accurate delivery date on your order confirmation after you ordered.
For more details, please ask to : email@example.com
These are stainless steel iris diaphragms which can change it's aperture size. It is mainly used for optical axis alignment of the laser beam and blocking the return light or stray light.
◦ It also can be used in a UV optical system or in a clean room.
◦ By sliding the adjustment lever, 12pcs diaphragm blades can be controlled and aperture will have a shape similar to a circular shape. By turning the adjustment lever, the aperture diameter can be fixed.
◦ The scale for aperture diameter is in the front of the holder.
◦ To minimize the outgassing, venting hole and grooving are provided onto screw holes and inset sections.
▶Diaphragm blades of vacuum iris diaphragm acts like a mirror because of no black anodized. Therefore if laser beam is irradiated, a part of incident light will reflect and return to the optical path.
▶The iris diaphragm is a very delicate mechanism. Pushing or pulling the blades cause malfunction and it cannot be recovered.
▶Heat from high power lasers may deform the blades and cause them to seize.
▶Please make sure that open the package in a clean environment and do not touch it with your bare hands.
|Primary material||Stainless steel|